Qualifying Probe Tips Probe Compensation
CMMs generally gather their data by touching the workpiece with a probe (either a solid probe or an electronic touch trigger probe) attached the machine's measuring axis. Although the tip of the probe is very accurate, once the probe is attached to the CMM, the location of the tip to the machine's coordinate system must be determined prior to measuring. Since it's the tip's circumference that touches the part, the probe's center and radius are determined by measuring a very accurate sphere (requalification sphere).
Once the center and radius of the tip are known, when the probe contacts a workpiece, the coordinates of the tip are mathematically "offset" by the tip's radius to the tip's actual point of contact (Figure 14). The direction of the offset is automatically determined by the alignment procedure.
We do a similar procedure when we park a car. The better we can estimate our offset from the exterior of the car, the closer we can park it to the curb.
|© 2013 Hexagon Metrology is part of Hexagon | Privacy & Terms | Site Map|
Hexagon Metrology offers a comprehensive range of products and services for all industrial metrology applications in sectors such as automotive, aerospace, energy and medical. By empowering our customers to fully control their manufacturing processes, we enhance the quality of products and increase efficiency in manufacturing plants around the world.